Should I remove runs that have been imported with warnings?
Data problems that don't prevent the run being imported will result in a warning status. If this happens, the run's ion map will be shown with a yellow outline in the list of imported runs. When selected, a list of warnings is shown in a yellow box at the right on the main ion map:
The yellow outline and warning icon on an imported run that has warnings
An example of the warning messages shown when the run is selected
Some examples of messages you may see are:
- 1328 scans after scan 3600 were ignored because they had lower retention times
- Some m/z-intensity pairs were skipped because the m/z values were lower than preceding pairs
In the future, we expect that enhancements to the support for different data formats will mean that further diagnostic information will be provided in these warnings. Keep an eye on the Nonlinear Dynamics blog and the data formats FAQ to monitor the availability of enhanced data import support.
"1328 scans after scan 3600 were ignored because they had lower retention times."
In our experience, this is typically caused by one of the following two situations:
- The run file actually contains data for multiple runs
- If you're not confident that each run in the file covers the same retention time range, then it's probably best to remove the run from the experiment. This is because, otherwise, you could end up with the tail end of a later run appended to the whole of the first run in the file. Conversely, if you're confident the ion map shows only a single run, it may be ok to keep the run in the experiment.
- The run file has been created using a non-standard representation of retetion time
- This situation appears to arise from the use of some third party tools that convert vendor-specific file formats into mzXML.
"Some m/z-intensity pairs were skipped because the m/z values were lower than preceding pairs."
This is known to occur with some files generated from Orbitrap Elite machines. The warning message appears if some m/z values within a scan are slightly lower than preceding values. In this case the points with lower m/z values are skipped.
In this situation, it is unlikely that you would need to remove the run from your experiment, but it may be worth contacting your Thermo representative for further advice.